1 APPA N/N DIGITAL MULTIMETER INSTRUCTION MANUAL 2 TABLE OF CONTENTS 1. SAFETY 1 2. PRODUCT DESCRIPTION 2 3. PANELS . APPA N, N Conversion Type: (AC + DC) True RMS reading on AIR CONDITIONING mode. ACV subsp. + % + 8 d. Input Impedance: 10 m?// pF. Centenary Materials — We offer quality service and products: APPAN APPA APPA_Series Digit Multimeters.
|Published (Last):||14 June 2008|
|PDF File Size:||2.67 Mb|
|ePub File Size:||17.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
SAFETY Review the following safety precautions to avoid injury and prevent damage to this product or any products connected to it. To avoid potential hazards, use the product only as specified. These statements identify conditions or practices that could result in damage to the equipment appaa other property. These statements identify conditions or practices that could result in personal injury or loss of life.
Appw avoid fire hazard, use only the fuse type and rating specified a;pa this product. Do not operate without covers. To avoid personal injury, do not apply any voltage or current to the product without the covers in place. Never apply a voltage to a connector on the product that is outside the range specified for that connector.
To avoid injury or loss of life, do not connect or disconnect probes or test leads while they are connected to a voltage source. To avoid electric shock, do not operate this product in wet or damp conditions. The following list provides a comparison of the features between meters. Black labels are appz initial settings, blue labels are selected by the xppa button.
LCD display with dual numeric readout. True RMS mode indicator. Low voltage resistance indicator. Maximum, Minimum and Average indicators. Main display unit indicators. Buzzer appa once for every key-press. Buzzer beeps twice for every invalid key-press. Zero at center graph zoomed 2. The beeper sounds with each new update. Every reading is subtracted by reference in memory in this mode. When store is full, Every press beeps twice. RANGE function is only enable in just data log mode. Any position measuring function change escaps out without storing any data to memory.
Pause time is seconds, exceed pause time stores as seconds. Pause and Log Rate setting quantities are 3. Auto Power Off function is disabled. Press Range key again to escape: Scrolls up data of function selected: Scrolls down data of function selected. Remove all power from the circuit before connecting the test leads.
Limit large current measurements 10 to 20A to 30 seconds and allow two minutes of cooling between measurements. Remove all power from the circuit and discharge Capacitor to be measured before connecting the test leads. If an open fuse is detected, ProbE is displayed and beep sounds continuously when the correct function position and probe insertion are applied.
Disable backlit auto off. Set temperature measurement default at F. Clear memory area for store. The meter turns itself off within 30 minutes if no controls or settings are changed. Restore power by switching apps. Functions with count mode divide the [ number of least digits ] by 10 when on count mode, in addition DCV to add 2 more digits on every ranges.
Specifications exclude under 0. Equipped with High Energy Fuse. For best measurements, with mode on nf ranges. Multiply the digit accuracy by 2 for F. Indoor Apps 2, m ft.
APPA N/N DIGITAL MULTIMETER INSTRUCTION MANUAL – PDF
Distribution level mains, fixed installation. Local level mains, appliances, portable equipment. Signal level, special equipment or parts of equipment, telecommunication, electronics.
Do not operate in environments where conductive pollutants may be present. Compliance was demonstarted to the following specifications as listed in the official Journal of the European Communites: Radiated and Conducted Emissions. The meter is not waterproof.
Do not expose the LCD display to direct sunlight for long period of time. To avoid damage to the meter, do not expose it to sprays, liquids, or solvents. Clean the exterior of the meter by removing dust with a lint-free cloth. Use care to Avoid scratching the clear plastic display filter. For further cleaning, use a soft cloth or paper towel dampened with water.
To avoid damage to the surface of the meter, do not use abrasive or chemical cleaning agents. Disconnect the test leads from any circuit under test and turn off meter. Remove the test leads from meter. Loosen the plastic screws on the battery cover.
Remove battery cover from case bottom. Install a new battery after removing the original one. Assemble battery cover onto bottom case with screw driver. Follow step 1 to step 4 described in Battery Replacement.
Remove the battery from meter. Remove 3 screws installed the top case and bottom case of meter. Separate the top case and PCB of meter.
Replace a new fuse fuse 1 or FUSE 2. Assemble the top case and PCB of meter. Assemble the top case, PCB, and bottom case of meter. Install the battery removed and assemble the battery cover.
Installing improper fuses can cause injury and product damage. Precautions and ap;a information Specifications Basic maintenance. Precautions and safety information Specifications.
Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Safety Information W To avoid damage to the meter, electric shock or injury: Never use the meter if the meter or test leads look damaged.
Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product.! Safety notes These meters have been produced to comply with category II requirements.
Precautions and safety information Specifications Performance test procedure Calibration. Specifications are subject to change without notice. All product names are trademarks of their respective.
Caution, risk of danger. General instructions This instrument complies with CAT. III V overvoltage standards. To get the best service. Silicone test leads 4 mm probes. One pair of red and black test leads One pair of Lifetime Limited Warranty Each Fluke.
APPA N INSTRUCTION MANUAL Pdf Download.
EC Declaration of Conformity Safety Information Use the tester only as specified in this Instruction Sheet, otherwise the protection provided by the tester may be.
Digital Multimeter – 7 sppa MOdel No: V3 Thank you for purchasing a Sealey product. Manufactured to a high standard, this product will, if used according to these instructions.
The Model provides three test ranges plus continuity. This device can measure Earth Ground. This user s manual describes the specifications and handling precaution for this.